The 1108 is a Hall-effect sensor that provides a voltage output that is proportional to the applied magnetic field.

#### Product Specifications

Sensor Properties
Sensor Type Magnetic Field
Controlled By VoltageRatio Input
Sensor Output Type Ratiometric
Magnetic Flux Min -500 G
Magnetic Flux Max 500 G
Measurement Error Max ± 0.5 %
Magnetic Zero Balance Max 15 G
Magnetic Resolution 1 G
Electrical Properties
Current Consumption Max 2 mA
Output Impedance 1 kΩ
Output Voltage Min 200 mV DC
Output Voltage Max 4.7 V DC
Supply Voltage Min 4.5 V DC
Supply Voltage Max 5.5 V DC
Physical Properties
Operating Temperature Min -20 °C
Operating Temperature Max 85 °C
Neodymium Magnets
Mangetic Field Strength (@ 6.5 mm) 500 G
Length 8.7 mm
Diameter 5 mm

## Getting Started

Welcome to the 1108 user guide! In order to get started, make sure you have the following hardware on hand:

Next, you will need to connect the pieces:

1. Connect the 1108 to the HUB0000 with the Phidget cable.
2. Connect the HUB0000 to your computer with the USB cable.

Now that you have everything together, let's start using the 1108!

## Using the 1108

### Phidget Control Panel

In order to demonstrate the functionality of the 1108, we will connect it to the HUB0000, and then run an example using the Phidget Control Panel on a Windows machine.

The Phidget Control Panel is available for use on both macOS and Windows machines. If you would like to follow along, first take a look at the getting started guide for your operating system:

### First Look

After plugging in the 1108 into the HUB0000, and the HUB0000 into your computer, open the Phidget Control Panel. You will see something like this:

The Phidget Control Panel will list all connected Phidgets and associated objects, as well as the following information:

• Serial number: allows you to differentiate between similar Phidgets.
• Channel: allows you to differentiate between similar objects on a Phidget.
• Version number: corresponds to the firmware version your Phidget is running. If your Phidget is listed in red, your firmware is out of date. Update the firmware by double-clicking the entry.

The Phidget Control Panel can also be used to test your device. Double-clicking on an object will open an example.

### Voltage Ratio Input

Double-click on a Voltage Ratio Input object in order to run the example:

General information about the selected object will be displayed at the top of the window. You can also experiment with the following functionality:

• Modify the change trigger and/or data interval value by dragging the sliders. For more information on these settings, see the data interval/change trigger page.
• Select the 1108 from the Sensor Type drop-down menu. The example will now convert the voltage into Gauss automatically. Converting the voltage to Gauss is not specific to this example, it is handled by the Phidget libraries, with functions you have access to when you begin developing!

## Technical Details

### General

The 1108 is optimized, sensitive, and temperature-stable. It is a ratiometric Hall effect sensor which provides a voltage output that is proportional to the applied magnetic field. Accuracy can be improved on a specific device with offset and/or gain calibration (using known flux).

### Formulas

The Phidget libraries can automatically convert sensor voltage into Gauss by selecting the appropriate SensorType. See the Phidget22 API for more details. The Formula to translate voltage ratio into Gauss is:

### Phidget Cable

The Phidget Cable is a 3-pin, 0.100 inch pitch locking connector. Pictured here is a plug with the connections labelled. The connectors are commonly available - refer to the Analog Input Primer for manufacturer part numbers.

# 1108_0 - Magnetic Sensor

• Brands Phidgets
• Product Code: ES000046
• Availability: In Stock